Signal level meter, TV meter, CATV meter
D202 Digital Signal level meter
5MHz to 870MHz
Radiasun is the leading manufacturer of cable and satellite meters in the world. Radiasun delivers the most satisfying products and services to the customers all over the world, with our own brands name, and also provides OEM and ODM for the most famous brands in the field.
5MHz to 870MHz
650nm to 1650nm
5(46) to 870MHz,
5(46) to 870MHz,
Welcome to Radiasun website.
We make cable TV signal level meter, satellite finder, optical power meter, PON optical power meter, optical light source, OTDR, fusion splicer, ADSL tester, IPTV tester, visual fault locator, etc...
Please send a quick message to us if you have any questions. We will reply you as soon as possible.
TV meter is also named TV signal level meter, field strength meter; For satellite TV it is named satellite finder, satellite locator.
TV meter is used to evaluate, test the quality of radio and television signals as measuring instrument. This article describes the basic principles of TV meter, key features and commonly used measurements.
TV meter is commonly used in radio and television equipment in broadcasting. It is mainly used in terrestrial television, cable TV, satellite TV signal measurement and analysis. It is always an important job for its systematic evaluation and measurement to radio and television project, but the TV signal strength is the most basic measurement. Through the field strength measurements, we can have a preliminary understanding to the TV system equipment, the quality status of a network operation.
1. Basic concepts of signal level meter
Signal level meter (field strength meter) is an instrument measuring TV signal strength. Strength stands for electric field strength; it is the level of signal that antenna sensing at one point in space, to characterize the electric field strength at this point. TV signal strength measurement is actually measured by a standard antenna to receive induction territorial television signals, or directly connected to the output of the cable TV signal and read the signal strength with level meter.
Signal level meter makes dB as the unit, such as dBuV, dBmV, dBm.
Optical power meter used to measure absolute optical power or measure a relative loss of optical power through optical. In optical system, measuring optical power is the most basic thing, is very like the multimeter in electronics. In optical measurements, optical power meter is commonly used in heavy duty work. A optical power meter is able to evaluate the performance of fiber optic transmission equipment by measuring absolute power of optical transmitter unit or optical internet. It is able to measure connection loss, test continuity and help to evaluate transmission quality of fiber link with a power meter and stabilized light source used in combination.
The unit of optical power meter is dbm. In the manual of fiber-optic transceiver or switchboard, has its light-emitting and receiving optical power. Usually, when the emitting is less than 0dbm, the minimum optical power that receiver received is called sensitivity; and the unit of the value that the maximum the receiver received minus the sensitivity is db (dbm-dbm=db), which is known as dynamic range. Emitting power minus received sensitivity is the allowed fiber attenuation values. The value of the optical power that actually emitted minus the actually received is the fiber attenuation (db). The optimum value of optical power that receiver received is “the maximum optical power that could receive – (Dynamic Range /2)”, but generally not so good, because the dynamic range of every optical transceivers and modules is......
2 Splicing Parameter Setting-fusion splicer
Turn on the power of fusion splicer and then the instrument begin self-checking. If everything was ready, fusion splicer mode would be entered, the display on the screen is as figure 2-2.
2.1 Main Menu-fusion splicer
Enter The Main Menu of fusion splicer
In the fusion splice mode, press the key and the instrument will enter the menu mode. The screen displays the main menu as...
4 Fiber Operation-fusion splicer
Step 1-fusion splicer: Before splicing, thread the fiber through the optical fiber protection sleeve. Step 2-fusion splicer: Remove fiber coating, and then brush the bare fiber with alcohol-soaked cotton swab. The typical bare fiber length is 20mm.The concentration of the alcohol should heavier than 905. Step 3-fusion splicer: Cut the bare fiber using the cleaver. The typical value of the cleaver length is 10mm~16mm. Step 4-fusion splicer: Open the shield and fiber holder clamp carefully. Set the fiber in the V-groove and the fiber holder on the holder mount. At the same time, note that the tip of the bare fiber does not touch anything. Make sure fiber is at the bottom of the V-groove. If not, reset the fiber. Step 5-fusion splicer: Set the other fiber in the same way. Put the shield down carefully.